The Elcometer range of digital coating thickness gauges has been specifically designed to provide highly accurate, reliable and repeatable coating thickness. The Elcometer Coating Thickness. Gauge is an incredibly versatile gauge. With a range of probes in both Integral or Separate probe versions for coating. Elcometer Coating Thickness Gauge. Operating Instructions. English. R. English Equipment described in these instructions is covered by the following.
|Genre:||Health and Food|
|Published (Last):||7 July 2011|
|PDF File Size:||6.46 Mb|
|ePub File Size:||15.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
Elcometer Coating Thickness Gauge Probe Range A range of fully interchangeable probes for the Elcometer separate coating thickness gauge. Follow the instructions ekcometer by the battery manufacturer when charging and disposing of rechargeable batteries. All other trademarks acknowledged. For gauge checks and certification contact Elcometer or your local Elcometer supplier. Remove the batteries elccometer the gauge if it is to remain unused for a long period of time.
Elcometer Coating Thickness Gauge A hand-held gauge for fast and accurate measurement of coatings thickness. Husqvarna Chainsaw Service Manual Re-calibrating using the smooth surface calibration method see Calibration Adjustment for Smooth Surfaces on page 17 to set zero will remove from the display.
Order using sales part number TEST To disconnect a probe, pull on the outer knurled sleeve of the plug – do not twist. The printer is off line, out of paper or printer or 3345 fault.
Postage cost can’t be calculated. Immediate payment required for this item.
The display will blinkc and then display the coating thickness. Elcometer User Manual for Mac To delete the last reading from elconeter statistical calculations when the statistics feature is switched on, press and hold for more than 2 seconds but less than 4 seconds. Protective Coating Inspection Kits This straight probe is specifically designed to measure the coating on both flat and curved Download Instruction Manual.
Report item – opens in a new window or tab. The backlight is designed for low light operations and can be turned on and off via the keypad. Marshall Manual – Manual The gauge displays the symbol of the first Statistics item on the statistics menu with its numerical position in the user list shown in the top left of the display. With the gauge switched on, press. The Elcometer Coating Thickness Gauge elcoemter powered by internal dry cell batteries.
The principle of electromagnetic induction is used for non-magnetic coatings on magnetic substrates such as steel. To fit or replace batteries: Selecting OFF and pressing6. Press or for more than 4 seconds to adjust the display to the calibration foil value, When close to the correct reading, a few quick presses of or will step to the correct reading.
Elcometer 345 Manual
The key-lock feature requires the keys used to access these routines to be pressed for 4 seconds before any response is made. Regular calibration checks elco,eter a requirement of quality management procedures e. To switch off, press and hold The gauge will switch off automatically after one minute of inactivity.
The eddy current principle is used for non-conductive coatings on non-ferrous metals substrates. Switch off the gauge. The reading may be inaccurate if the probe is not held flat against the surface.
Confirm that the average gauge reading on each calibration foil is correct. The bleeper will sound to indicate the instruction has been accepted. Press and hold for 4 seconds to set the display to zero.
Your gauge was calibrated at the factory, however you can adjust the calibration using the set of calibration foils supplied with the gauge. This will cause the gauge to display 1 Err. To change units again, repeat these steps. Release to delete eclometer last reading, Note: This may be due to incorrect substrate, incorrect values or units, etc. When the elcometr is shipped from the factory it comes with a preset calibration.
Rechargeable batteries can be used if they are charged outside the gauge. The calibration values are stored in the gauge until elco,eter values are entered, even when the gauge is switched off. Separate probes can be fitted by the user. The Elcometer Coating Thickness Gauge is a world beating product. When switched on, the backlight comes on when a reading is taken and remains on for 3 seconds. Lift off battery compartment cover at rear of gauge.
Press and hold 5.
for sale paint gauge , elcometer
Example of Statistical User List. If the display is heated above 50C F it may be damaged. Switch the gauge on and pressthe symbol in the display will flash. If looked after, it will last a lifetime. Line up the 2 red marks when inserting the plug. Place the probe on the bare metal, wait for a reading to be displayed. Place the chosen thin calibration foil onto the bare metal surface.